- Resonance frequency analysis of thermal acid-etched, hydrophilic implants during first 3 months of healing and osseointegration in an early-loading protocol
- International Journal of Oral & Maxillofacial Implants
- Volume | Issue number
- 30 | 4
- Pages (from-to)
- Document type
- Faculty of Dentistry (ACTA)
Purpose: Safe loading of dental implants requires an optimal osseointegration. This osseointegration process during healing could be analyzed by resonance frequency analysis (RFA). The purpose of the study was to evaluate RFA changes during healing in splinted, early-loaded, thermal acid-etched, hydrophilic implants over time. Materials and Methods: Patients received a minimum of two implants: an implant with the prosthetic abutment connection at the crestal bone level (bone level) and one with the prosthetic abutment connection at a 2.5-mm supracrestal site (tissue level). Implant stability was measured at weeks 0, 2, 3, and 12 using the Osstell device. Results: Seventy-six implants were placed in 32 patients. By week 2, early-loaded tissue-level implants showed a significant drop in mean ± standard deviation (SD) implant stability quotient (ISQ) values of 2.2 ± 3.6 (P < .001). Changes in ISQ values were significant between weeks 3 and 12 and also between weeks 0 and 12, with mean differences of 4.2 (P < .001) and 2.8 (P < .001), respectively. Early-loaded bone-level implants show a significant change in ISQ of 2.3 ± 3.7 at week 2 (P < .01) and -1.3 ± 4.7 at week 12 when compared to an ISQ value of 2.9 ± 4.9 at week 3 (P < .01). Bone-level implants achieved higher ISQ values compared with tissue-level implants at weeks 0, 2, 3, and 12, with mean differences being 3.8 ± 5.5 (P < .01), 3.8 ± 6.1 (P < .01), 3.7 ± 6.7 (P < .01), and 2.3 ± 5.8 (P < .05), respectively. Conclusion: This study found a significant dip in ISQ values, with the lowest point seen at week 2. ISQ values remained higher in bone-level implants throughout the process of healing and osseointegration.
- go to publisher's site
If you believe that digital publication of certain material infringes any of your rights or (privacy) interests, please let the Library know, stating your reasons. In case of a legitimate complaint, the Library will make the material inaccessible and/or remove it from the website. Please Ask the Library, or send a letter to: Library of the University of Amsterdam, Secretariat, Singel 425, 1012 WP Amsterdam, The Netherlands. You will be contacted as soon as possible.