C.M. ten Bruggenkate
- A 5-year prospective multicenter study of early loaded titanium implants with a sandblasted and acid-etched surface
- International Journal of Oral & Maxillofacial Implants
- Volume | Issue number
- 26 | 6
- Pages (from-to)
- Document type
- Faculty of Dentistry (ACTA)
For dental implants to be successful, osseointegration must occur, but it is unknown how much time must pass for osseointegration to be established. Preclinical studies suggested that titanium implants with a sandblasted and acid-etched (SLA) surface were more osteoconductive and allowed more rapid osseointegration than machined or turned implant surfaces. The hypothesis of this study was that implants with an SLA surface could be loaded in half the conventional healing time of machined-surface implants and that, after loading, the implants would be successful for 5 years.
MATERIALS AND METHODS:
A prospective multicenter clinical study was conducted with 439 implants placed in native bone in 135 edentulous and partially edentulous patients. Abutments were attached to the implant with 35 Ncm of torque without countertorque after 6 weeks in type I to III bone and after 12 weeks in type IV bone. The patients were carefully evaluated for 5 years.
Most implants were placed in nonsmoking, nondiabetic patients with a mean age of 55 years (range, 21 to 82 years). Eighty percent of the implants were 10 or 12 mm long, 96% had a diameter of 4.1 mm, and 78% were placed in type II or III bone. Patients maintained good oral hygiene and were satisfied with the restorations. Four implants failed, and one implant was deemed unsuccessful between surgery and the 1-year postloading visit. No implants failed or were unsuccessful in subsequent years. The cumulative survival and success rates for 385 implants in 120 patients after 5 years were 99.1% and 98.8%, respectively.
Implants with an SLA surface can be restored in 6 weeks for type I to III bone and 12 weeks for type IV bone. Furthermore, they can be maintained after loading for 5 years with very high success and survival rates.
If you believe that digital publication of certain material infringes any of your rights or (privacy) interests, please let the Library know, stating your reasons. In case of a legitimate complaint, the Library will make the material inaccessible and/or remove it from the website. Please Ask the Library, or send a letter to: Library of the University of Amsterdam, Secretariat, Singel 425, 1012 WP Amsterdam, The Netherlands. You will be contacted as soon as possible.